View and Download HP A calibration manual online. Synthesizer/Function Synthesizer HP A Operating And Service Manual. Frequency ( pages) . The Operating and Service Manual contains infor¬ mation required to install, operate, test, adjust, and serv¬ ice the Hewlett-Packard Model A. Find great deals for HP a Function Generator Operating and Service Manual . Shop with confidence on eBay!.
|Published (Last):||25 February 2013|
|PDF File Size:||7.21 Mb|
|ePub File Size:||9.43 Mb|
|Price:||Free* [*Free Regsitration Required]|
In Unconfiguring, the controller negates the bit and level assignments for all or selected 3325z. Instruments with Option do not have rear panel signal output.
Seal the shipping container securely. Program examples are shown in Appendix B-3, using the -hp- Model A Calculator as the system controller. Line 5 Return to Line msnual to continue sweeping. Set the A to the following frequencies and verify that all harmonics are 65dB below the fundamental. Otherwise, both kits are available separately by their -hp- part numbers. Unrecognizable data character 9.
HP A CALIBRATION MANUAL Pdf Download.
Without changing the reference level, change the spectrum analyzer center frequency to Set the dc offset to the positive and negative voltages shown below. Included are initial inspection procedures, power and grounding requirements, line voltage selection, environmental requirements, installation instructions, HP—IB connection procedure, and instructions for repackaging for shipment.
Displays the value of the parameter selected, error codes, failure modes, HP-IB address, amplitude ana phase modulation state. The digital voltmeter reading should be —4, to —4. Change A amplitude to. Verify that all three readings are between. Keyboard and Display, A Table briefly summarizes the engineering effort that has brought A6 to its current revision. The following general instructions should be used for repackaging with commercially available materials.
Repeat steps d and g.
Placing the A in Remote. You can help improve this publication.
3325A Synthesizer/Function Generator
Change A function to Triangle. Press the High Voltage Output key again to turn the option off. See Service Note A-5A for a modification procedure to improve square wave phase control in these instruments. This also applies for Figure New test limits are as follows: Oscilloscope display should show one square wave per division, 5 divisions peak-to-peak vertical.
HP a Function Generator Operating and Service Manual | eBay
Repeat Step d at 2 kHz and. Page 20 TB 4 Enter Polling – Polling is a means servics which a controller can identify a device that needs interaction with it.
For a negative phase shift, press the key before entering the numerical data. These instruments contain the de offset control circuitry shown in Figure Output Level and Attenuator Check.
HP 3325a Function Generator Operating and Service Manual
The following examples are given to illustrate the basic procedure for developing a program. This number can be used to order 4×6 inch transparencies of the Servicr and Service Sdrvice, Each Microfiche contains up to 96 photo-duplicates of the manual pages. Maximum and minimum amplitudes for each function are shown in Table Change the power supply output to the following voltages and repeat Steps j through m for each. At 18MHz, check for the 3: Box 69, Marysville, Washington Printed: Failure to allow adequate air circulation will result in excessive internal temperature, reducing instrument reliability.
Include instrument model and serial numbers.